:: 6.35mm Hex
:: 4.0mm Square Adapter for other systems is included!
:: Compatible with: Alpha Bio®, Nobal Biocare® and many other systems
The wrenches are not sterilized prior to shipping.
They must be cleaned, disinfected and sterilized before use.
When cleaning, do not use abrasive detergents and do not apply excessive pressure.
To avoid damage do not clean with wire brush or metal pads.
Use only cleaning and disinfectant solutions with a ph between 5 and 10.
Please follow the cleaning/disinfectantmanufacturer’s instructions
(e.g. intended use, dosage, time, renewal period).
Frequent but careful cleaning has little effect on the life of the ratchet.
Product life is determined by wear.
Recalibrate the wrench once a year. The wrench must be cleaned and sterilized after service.
Recommended Sterilization Method
Loosen the ratchet completely prior to sterilization
Method Fractionated pre-vacuum process (according to ISO 17665)
Temperature Heat up to 134⁰C (137⁰C Maximum)
Pressure 3 pre-vacuum phases with at least 60 millibar pressure
Holding Time at least 5 minutes
Drying Time at least 10 minutes
After sterilization check the packaging of the sterilized instruments for damage. Check the sterilization indicators.
Store the sterilized wrenches in a dry, dust-free, corrosive vapor-free area at a moderate temperature.